The unit is used under the sole responsibility of the user without liability to the manufacturer. We assume no liability for damage or loss resulting from misuse, mishandling of any Medical Frequency Program – BioPulse v 3.0 Software component. We make no guarantees or warranties with respect to the contents or the use of this manual and/or Medical Frequency Program – BioPulse v 3.0 software unit.
The information provided in this manual is for informational purposes only and is not intended as a substitute for advice from your physician or other health care professional. You should not use the information in this manual for diagnosis or treatment of any health problem or for prescription of any medication or other treatment. You should consult with a health care professional before starting any diet, exercise or supplementation program, before taking any medication, or if you have or suspect you might have a health problem. You should not stop taking any medication without first consulting your physician
Asociacija Alternatyvi Medicina grants Buyer non-exclusive right to install and use Medical Frequency Program – BioPulse v 3.0 on a single computer and a single terminal for the purpose of operating products purchased by Buyer from Asociacija Alternatyvi Medicina.
You may install one additional copy on a laptop or a home computer provided that it is owned by you, for your use for archival purposes.
The Software’s content is protected by copyright laws. Buyer may not rent, lease, loan the Software without the prior written notice of Asociacija Alternatyvi Medicina. Buyer may not disassemble, copy, reverse engineer, decompile, modify, sublicense, assign, create a derivative work from, or use the Software in any manner except as stated in this Agreement.
We do not warrant that Buyer’s use of the Software will be error-free and uninterrupted. In case of defective Software due to workmanship, Buyer may return a faulty disc along with dated proof
of purchase to correct any error or replace the defective disc free of charge.
Except as set forth in this paragraph, the Software is licensed “As Is” without warranty of any kind, expressed or implied, by statute or otherwise, including but not limited to the implied warranties of merchantability, non-infringement and fitness for any particular purpose. We do not guarantee or warrant the suitability of the Software or that it will meet your requirements.
The liability of Asociacija Alternatyvi Medicina under the warranty set forth above shall be limited to the amount paid by Buyer for the product.
Asociacija Alternatyvi Medicina and its suppliers will not be responsible for any indirect, incidental, special, or consequential damages (including damages for loss of business, loss of profits, etc.) whether based on breach of contract, tort and negligence, product liability and the like Asociacija Alternatyvi Medicina liability to Buyer will be limited to the amount paid for the product.
Electric cardiac pacemaker; coagulation failure; congenital deformity of central nervous system; implantable cardio stimulator; individual intolerance to current electricity; injury on a skin after electro stimulation; state of acute psychic excitement or intoxication (alcohol), pregancy.
It is necessary to remove ornaments, watches, glasses. The clothes should be from natural fabric. Remove all devices radiating electromagnetic waves such as beeper, cell-phone etc. minimum on a meter.
Time of frequencies generation can vary, it depends on the quantity of frequencies recorded in the device and preset time. You can set any time for a frequency. Usually 3-4 minutes are enough for generating of one frequency, but in some cases see the commentary to the disease and frequencies.
Quantity of sessions is selected individually on actual state of health.
At preventive care it is recommended to perform 10 sessions every other day, with an interval 1-3 months. When you treat diseases, the sessions are carried out each day, 1-2 sessions per day. Quantity of procedures -10-15 or more, then make an interval for 1-2 weeks.
The device connection/disconnection is realized only when the unit is switched off.
The connection of the USB- port cable, electrodes for frequency correction is realized only in dedicated for them jacks (sockets).
The computer should be reliably grounded.
The present Maintenance Guide is intended for acquaintance with the rules and order of frequency therapy and Medical Frequency Program – Biopulse v 3.0 software operation.
Medical Frequency Program – Biopulse v 3.0 software has been designed to be easy and simple to use and comes with a comprehensive guidance to maximize ease of use. The Maintenance Guide contains information necessary for correct software operation and maintenance. You should read the manual carefully before using your software.
Software contains the health correction module with frequencies.
There are several buttons helping you to destroy pathologic frequencies and strengthen your physiological frequencies in an organism.
The influence on a person is performed in a contact way with the metal electrodes (which are supplied together with the software).
Before installing the software make sure that the computer satisfies with the technical requirements.
Put CD with “Medical Frequency Program - Biopulse v.3.0” software into CD-ROM drive.
You need to copy the software from CD on the hard disk of the computer. Open the folder and you will see there a file . Click on this file and installation process begins.
You need to press to choose a directory where you want to install the softwareto continue with installing the software. The next window creates a shortcut icon of the software in the directory. You can also rename the folder.
Then press the button again
The software is ready to be installed on your computer and setup program informs you about it.
If you agree with the setting you chose, press the button to proceed with the installation.
When the software is successfully installed on your computer you will see the following window:
You need to press the button
The software Medical Frequency Program – Biopulse v.3.0 is successfully installed on your computer
Before starting the device software make sure that the unit is connected to the computer and is switched on. Check the unit switching on by turning on the power button on the device side panel. A short sound signal is generated if the device is on.
Check the connection of the device, electrodes for frequency therapy, USB-port cable to the device and computer visually.
To start the program, click the program .
Main window of the program will appear on the screen.
After the start of the program the Introductory window appears on the screen. Some elements are typical for all windows in Windows, some are specific only for the device’s software.
The Title bar is on the top of the program window, the name of the program (Medical Frequency Program – BioPulse v 3.0) is indicated in Title bar. The title bar is always highlighted by the blue color that testifies the activity of the given window and possibility of work with the given window at the moment.
With the window control buttons which are on the right in the title bar you can: minimize the window into a button on a taskbar , to maximize the window on full desktop , to restore the window to its previous size , to close the window .
Click File -> Open Project - use this button to open saved projects of frequencies directly from Toolbox panel that you saved for further usage.
If you want you can create you own projects consisting of the frequencies, so that to use the same frequencies next time 2 buttons will be helpful:
Click File -> Save Project - this button saves the frequencies you want to use next times
helps you to choose the time of frequencies generation. You may enter necessary frequency, time and wave form manually.
Also you can use this field for creating your own set of frequencies as a project and save it in a certain directory on your computer.
Left or right click of the mouse on the figures will help you to choose a necessary frequency, time of generating and wave form, then press add, the frequency will appear in the field Current list
Then press the button to keep the frequencies in the chosen directory on your computer for further usage. Or alternatively press the button Save Project from the toolbox panel.
Click File -> Print - to print the information out (frequency name, frequency comment, list of the frequencies for a particular disease).
electrode on solar plexus, one on belly. cafl
Click File -> Exit - use File and Exit tabs to exit the program.
Click Description -> Add
You may enter in the field Name the name of disease or problem.
Commentary - the notes about frequency
Frequency List –
left or right click of the mouse on the figures will help you to choose the frequency value you want to in the list and then click
Alternatively you may the frequency or the frequencies from Frequency List and operation.
Choose from the Description List a disease, click Description -> Edit
and edit information the disease contains.
The buttons will help you to manage your records.
Click Description -> Delete – to delete the disease from the software’s database.
The software will ask you to confirm the deletion.
Click Description -> Insert To Current List
Choose necessary disease name from the list. This field displays the list of diseases in alphabetical order.
The vertical scrolling bar is on the right from the list of diseases.
Click the button on the scrolling bar to move the list of diseases correspondingly up or down.
click . Choose a waveform and set a second, minute or hour for a frequency’s generation. Usually 4-5 minutes are enough, but see additional information in commentary field.
The commentary on the disease (some information that might be helpful) will be displayed in the middle window together with the frequency value itself.
If there will be “XXXX” signs instead of figures, they are the frequencies which are encoded in the disease name and they will be generated as usually.
- searches the necessary disease from the list. You can search it by key word in its name (description), comments or if you remember the frequency associated with the given disease.
Check the box with description if you want to find ‘Energy and vitality’ and enter the name in the search field
or use the comments about this name if any or a frequency in your search
Click Description -> Print - to print out the disease’s description.
The print list will look in the following way:
There can be printed several diseases that you choose from the list of diseases.
Alternatively you may use the buttons located under the field Description List to execute the operations on adding, editing, deletion, printing and insertion.
Click Description -> View
This function enables you to display the diseases according to alphabetical order, to view the diseases starting form a certain letter or hide all names.
Press Description -> View -> Select All and all diseases in the software’s database will be displayed in alphabetical order.
If you want to be displayed only the diseases starting from “A” letter, press Description -> View -> Unselect All check “A”
Click Description -> View -> Invert Select.
You may check a certain letter so that it won’t be displayed in the list together with the rest diseases or you may select the all other letters except the chosen one.
You click View, remove the check from the letter “C” and press Invert Select, the diseases starting from letter “C” will be displayed in the list.
Or Click View, remove the check from the letter “C’ and the list with the diseases will be given without “C” letter: A, B, … D, E, etc.
Left or right click of the mouse on the figures will help you to choose the frequency value, time of generation and wave form you want to in the list
The buttons from the or
from the Current List will help you to manage the software’s operation.
Click Actions -> Generate to choose the way of frequencies’ generation either through a software or device and electrodes
Where Sound Blaster is generating through the software and USB Device is generating through device “Biopulse TC”
Click Actions -> Sweep – to choose the function in the software
In the same way as in you may generate this option through software and device.
Sweep function is the function of generating the frequencies in a certain frequency range that you may set. As in our example, we want to generate the frequencies starting from 20 Hz till 1000 Hz with a frequency step 0,1 Hz. It means that the frequencies will be generated by default with an interval 0,1 Hz, the next frequency after 20 Hz will be 20,1 Hz , then 20,2 Hz and so on until it reaches 1000 Hz. You may also set the time for every frequency generating in the field Step period and select a waveform in Signal waveform. For example, 20 Hz will be generated within 3 seconds with the selected square waveform, then 20,1 Hz will be generated also within 3 seconds in the same waveform and so on until 1000 Hz.
You can also switch the inductor coil checking the box with .
The button saves the frequencies in the device.
Press the button to generate the frequencies sweep from the software.
Click Actions -> Read/Record - this function enables you to record chosen frequencies in the device’s memory to generate them independently from a computer (if you purchased the software with computer).
Click Info -> Help – click the button to get online assistance form Support Team
Click Info -> Credits
This function is not active yet. It will be available for credits intake.
Click Info -> Home Page and it will direct you to our website home page.
Record of the frequencies in the device:
1) Put the electrodes plug and USB-port connecting cable in dedicated for them jacks.
Note: If you use the same set of frequencies as the last time you recorded in the device, you don’t need to use data input PC connecting cable;
2) Turn ON/OFF button on the device (if you did not use the testing function before it);
3) Run the software Medical Frequency Program – BioPulse v.3.0
4) Choose the diseases from the list
and insert in the Current List
The frequencies marked with green color mean that they are ready to be inserted in the Generator List
Then press so that the frequencies will appear in Generator List
You can change the set of frequencies every time you want by choosing diseases from the list and adding them for generation in the toolbox panel
Usually you can change the one set of frequencies for another after 10-14 days of frequencies generation, when you complete with one set of diseases and you may begin with correction of another diseases. But you need to change periodically the frequencies’ wave form after 7-10 days of usage so that the organism does not get used to it.
The time of correction with frequencies depends on the quantity of frequencies you recorded scheduled.
For example if you set time for one frequency generating 3 minutes and there are totally 20 frequencies in the list, then you should seat with electrodes for one hour.
The running green line points that the process of frequencies record is performed.
When the frequencies are successfully recorded in the device the message will appear
The frequencies record in the device is performed with the USB-port connecting cable, it should be plugged into a special jack in the device and USB-port.
You can perform the frequencies generation in 2 ways: either through software or device.
Choose from the toolbox panel
and press so that the frequencies will be generated through the sound blaster of the computer (in this case you need to plug simple earphones in the headphones jack of the computer and put the earphones on any part of the body; do not put the earphones into ears).
Or you can generate the frequencies from the device managing the process through the software.
It is done in the following way:
Plug the electrodes for frequency therapy in the device’s jack. Connect the USB-port cable to the device and computer. Switch the device on, run the software.
Choose from the toolbox panel
and press the button
Take the cylindrical electrodes in left and right hands;
The frequency marked with green color means that it is generated at the current moment.
When the frequencies’ generation is completed the software gives you a message that the treatment process is completed.
ZAPPER function– this function starts the program for expulsion of worms, parasites.
The procedure of Zapper session:
Press Zapper, the frequencies are played during 7 minutes, then a short beep is given and you have to make a break within 20 minutes. After that press Zapper again and hold the electrodes during 7 minutes. Then again make a 20 minutes rest and play the last 7 minutes. The order of procedure is obligatory in this mode.
You can make such procedure once a week for preventive care.