Asociacija Alternatyvi Medicina grants Buyer non-exclusive right to install and use the Software
on a single computer and a single terminal for the purpose of operating products purchased by Buyer from Asociacija Alternatyvi Medicina. All rights in the Software not specifically granted in this Agreement.
You may install one additional copy on a laptop or a home computer provided that it is owned by you, for your use for archival purposes.
The Software’s content is protected by copyright laws. Buyer may not rent, lease, loan the Software without the prior written notice of Asociacija Alternatyvi Medicina. Buyer may not disassemble, copy, reverse engineer, decompile, modify, sublicense, assign, create a derivative work from, or use the Software in any manner except as stated in this Agreement.
We do not warrant that Buyer’s use of the Software will be error-free and uninterrupted. In case of defective Software operation, we perform technical after sales support.
Except as set forth in this paragraph, the Software is licensed “As Is” without warranty of any kind, expressed or implied, by statute or otherwise, including but not limited to the implied warranties of merchantability, non-infringement and fitness for any particular purpose. We do not guarantee or warrant the suitability of the Software or that it will meet your requirements.
The liability of Asociacija Alternatyvi Medicina under the warranty set forth above shall be limited to the amount paid by Buyer for the product.
Asociacija Alternatyvi Medicina and its suppliers will not be responsible for any indirect, incidental, special, or consequential damages (including damages for loss of business, loss of profits, etc.) whether based on breach of contract, tort and negligence, product liability and the like
Asociacija Alternatyvi Medicina liability to Buyer will be limited to the amount paid for the product.
All technical related issues should be contacted by Skype: thinkhowme
Thank you for purchasing the software Su-Jok 6 Ki and corresponding device. You are about to discover the comfort and convenience of our machine and software.
The Instruction Manual is to acquaint you with the rules and order of electro-acupuncture testing using the Su-Jok 6 Ki software The software has been designed to be easy and simple to use and comes with a comprehensive guide for maximize ease of use. The Instruction Manual contains information necessary for correct software operation and maintenance. You should read the manual carefully before using your software.
Our software can be used to assess a whole body health via via meridian measurements and obtain information for proper subsequent influence that you use in your practice. The software is intended for getting system evaluation, for defining functional activity of organs and systems with the help pf electrical conductance measurements and problems’ correction with color therapy inbuilt in the software and in the detecting probe.
The main objective of the software to become integrated with a computer in a more user-friendly way for a practitioner. It allows the practitioner to send results of reflexology zones measurements directly to their PC and create automated and accurate information for providing patients with an objective, patient-friendly, graphical print out at each visit. The direct reading from probe to computer saves the practitioner's time and allows him to obtain analysis information of the entire body within a short time.
1) Usage of the device is authorized only after acquaintance with the given instructions.
2) Use the device to avoid fire, electrocution or injury. Keep the device away from children. Always supervise children near the device. Store the device indoors in a dry location. Do not use the device in or nearby water.
3) Most of our devices are made of plastic and could melt or burn if they come in contact with any heat source or other fire. Do not set up, use or store devices against or near any open flames, heaters, furnaces or other hot objects and do not place it under direct sunlight.
The place of examination should be protected against the high-power sources of
electromagnetic fields to avoid the interference in the measurement process. If
there are other electrical devices which are the sources of electromagnetic
fields, it is necessary to switch them off before person’s examination.
The connection of the USB- port cable, electrodes for frequency correction, measuring active/passive probes and passive probe is realized only in dedicated for them jacks (sockets).
The computer should be reliably grounded.
An examinee should take off watches, rings, glasses with metallic items, jewelry which may interfere with the readings should be removed before examination. Hands and feet should not be moistened with oils, lotions or excessive perspiration. It is undesirable to take pills, alcohol, coffee affecting nervous system prior to the examination. Before taking a measurement wait for some time after drill, meal, etc. A person should be calm and relaxed. Don’t hold the passive electrode too firm or too loose as it can create altered measurements (too firm contact creates higher readings, too soft contact creates lower readings).
To create an accurate reading an examiner should not touch the examinee during
measurement (only touching the measurement point by the active/passive probe,
no physical contact).
Make sure that the moistened cotton on the active/passive probe is in contact with metal of the probe. The moist cotton should not be too wet or too dry. It is appropriate to douse in water the active/passive probe to maintain cotton moistened during the test.
If the skin of the examinee is too moistened with oils, lotions or perspiration dab gently excess moisture avoiding friction over the hands. During measurement don’t press too hard or too gentle on the examined points.
Electric cardiac pacemaker; coagulation failure; congenital deformity of central nervous system; implantable cardio stimulator; individual intolerance to current electricity; injury on a skin after electro stimulation; state of acute psychic excitement or intoxication (alcohol), pregnancy.
The unit is used under the sole responsibility of the user without liability to the manufacturer. We assume no liability for damage or loss resulting from misuse, mishandling of any software component. We make no guarantees or warranties with respect to the contents or the use of this manual and/or software.
The information provided in this manual is for informational purposes only and is not intended as a substitute for advice from your physician or other health care professional. You should not use the information in this manual for diagnosis or treatment of any health problem or for prescription of any medication or other treatment. You should consult with a health care professional before starting any diet, exercise or supplementation program, before taking any medication, or if you have or suspect you might have a health problem. You should not stop taking any medication without first consulting your physician
1) signal USB-port cable;
2) Su Jok 6 Ki installation software
3) Active/passive detecting tester (probe)
4) Passive electrode
Make sure that your computer meets the following requirements to avoid problems that will result from not having the following specifications:
Pentium III or above
The operating system Windows XP, Vista, Windows 7
RAM 512 Mb or above
100 Mb of free space on the hard disk
One free input/output USB-port
Attach the to the computer according to safety measures and general instructions in a following order:
· place the device and PC on the workplace
· turn the device and computer off before connecting device to computer
· after connecting the device to the computer connect the cable of the active/passive probe or the cable of the passive electrode to dedicated jacks, connect the USB port cable to the free input/output of the computer
Place and mount the device according to safety measures and general instructions of the present manual in a following order:
· place the device elements and PC on the workplace;
· connect USB port cable to the free input/output of the computer.
· Install the drivers copied from CD provided together with the device and software.
Put the CD with the software into CD-ROM drive.
To copy the software from the CD on the hard drive of the computer, locate the file. Open it with the left click of the mouse to begin the installation process.
To complete successful installation of the software on your computer, press the button Finish
Before running the software, make sure that the device is connected to the computer and switched on. Make sure that USB-cable is plugged into USB-port of the computer, and the other end of the cable to the device’s jack. A short sound signal is generated if the device is on.
To start the program, click on the software’s icon.
The Introductory window will appear on your screen.
If there is no information about patients in the database, you will see an empty “Clients” field. Before starting a test, add the patient to the database.
If client information is already in the database, it appears in the Introductory window of the client list.
If you want to exit from the program click the on the button, which is in the program’s Title Bar.
After the start of the program the Introductory window appears on the screen. Some elements are typical for all windows in Windows, some are specific for the device’s software
The Title bar is on the top of the program window, the name of the program is indicated in Title bar. The title bar is always highlighted by a green color. It indicates that the window is active at the given moment.
The window control buttons on the right in the title bar allow you to: minimize the window into a button on a taskbar , change the window’s size close it .
Toolbox panel is under the title bar
Languages - use this function to select the language of your software or translate to your native language directly in the software.
Settings – supposed for changing your proxy settings.
Info – provides general information about the software and also enables a user to read the manual online.
Help - get information about software and device operation. You are connected with our
Support Team through the Internet.
Status line is at the bottom of the Introductory window.
shows client’s name and surname.
- displays time and date of a test.- defines device’s number. In this example it shows that device is not connected to the computer.
Menu is in the left field of the software’s window. Press necessary button to execute the command.
The Clients tab is a management database, which records client information (such as name, last name, phone, etc.) and test histories.
Vertical scroll is right to the client’s card .
Press the buttons Up or Down on the vertical scroll to move a list with clients.
To choose a client from the list click on the client’s Name, Surname.
The line with the selected client will be highlighted in blue.
Review of the main functions in Client’s card menu
Buttons New, Edit, Deleteare used to work with the client’s information
Each time you want to view the information about a person, edit or delete it, choose the client by pressing his Name or Surname under the tab .
Click the button – to enter a new client’s card and type in the Name, Surname, Date of birth, Sex and contact details.
When you enter the information for a new client, the software’s window appears in a following way:
Client’s card has various fields. Press the necessary one and enter the all information required for it.
To upload a picture, press the button and browse the picture from file. To delete it, press
Click the button to keep the all information in the client’s card.
New client will appear in the list after that. To cancel the new client’s card, press .– changes the information in a client’s card.
Select the client’s name and press the ‘Edit’ button and make all necessary changes.
Click “Save” . All changes are saved in a client’s card. If you want to cancel the all changes, press the button .
– this button removes a client from the client’s list. Choose necessary client in a client’s card and press .
Then confirm or cancel the client’s deletion:
Note: once you delete the client, all his personal details are removed including measurements’ history.
This option enables a user to search for a necessary client by Name, Surname, Date of birth, Date of test.
Enter a text in a search line, mark a search criteria and press the button
Before you open the window with diagnostic function, you
need to choose a necessary client from your Clients' list or introduce a new
Measurement window appears on the screen.
It is necessary to start measurement of Su Jok points with the preparation of the device.
Check visually the connection of the
passive electrode, detecting probe, USB port cable.
Insert the moistened cotton tampon into the active/passive electrode’s search cup.
The software provides you with 2 types of
You can measure the points either through 5 elements or 6 Ki
Depending on the type you choose, the software displays different points locations.
If you decided to choose 5 elements testing, it is possible to test traditional 24 points or 120 points (5 points on each meridian).
Evaluation form for 5 elements testing displaying:>
24 points or 120 points including Left and Right sides for Yin and Yang
Evaluation form for 6 Ki testing including 24 points or 144 points
24 points or 144 points
including Left and Right sides for Yin and Yang
The software directs you in performing all 24 points (120 or 144 ) measurements. Every point is graphically illustrated where the measurement point located which a patient and practitioner may view and follow the screen. The pictures clearly show the precise location of the acupuncture point to examine, and it is simple even for new users.
The points which should be tested are marked with different colors in correspondence with 5 elements and the point which is tested at the moment is blinking.
Follow the points’ illustration to locate and complete all measurement points.
The values of skin conductance are located on the middle side of the Testing window.
The display consists of the numerical values of the skin conductance and a dynamic red line. The red line is a real time graphing of your skin conductance received from the organs’ representative points.
The display scale can change its color during measurement process. There are 3 colors for
preliminary evaluation of the measured point depending on the
electro-conductivity of the point and meridian connected with it: green, blue
and red. Blue color means low deficient value, green designates normal state of
the point and red color tells us about excessive values.
The vertical line on the left side of the graph displays numerical values of skin conductance level.
Depending on the value measurement field is marked with different colors enabling a user to evaluate the points preliminary (blue for deficient values, green for normal and red for excessive results).
The program automatically calculates and enters the average skin conductance level for the particular point measurement. All readings are displayed in the evaluation form under the display scale.
The software provides by default the function to shift automatically to another point measurement when you complete measuring previous point.
You can choose the time from 0 to 60 seconds, within this period the software will shift you to another point testing. For example, if you select 0 seconds, it shifts you to measurement of another point at once; if you select from 1 till 60 seconds, the program has to shift to another point within 1, 2, …..60 seconds depending on time chosen.
This field displays information about the point being tested currently, like in our example Meridian: Lung, Point – A III, Energy – Fire
The description and rules of the TESTING process.
The device is connected to the computer, USB-port cable is connected to the computer and device, active/passive probe, electrode are connected to the device. Run software.
Make all necessary general requirements to examination procedure.
Make a small size of cotton swab. Moisten it with water or salt solution and squeeze the cotton swab in the active/passive probe’s cup. Make sure that the moistened cotton on the active/passive probe is in contact with metal of the probe. The moist cotton should not be too wet or too dry. It is appropriate to douse in water the active/passive probe to maintain cotton moistened during the test.
Have the client hold the probe not too loose or not too tight.
With the feet resting on non-conductive ground the client seats comfortably on a chair or lying on a bed. The examinee should not contact metal, wall furnishings.
Without touching, probing or palpating, place the active/passive measuring probe with moistened cotton tampon over the examinee’s test point aсcording to the point’s illustration, since that moment the reading begins. The red line steadily rises to a certain value where it gradually stabilizes.
The measurement of each point is performed within 3 seconds and the program automatically gets three measurement readings of the meridian conductance value on each second, then the software calculates the average value of the point.
After 3 seconds when the running red line stabilized and reaches the end of display scale, measurement value is accompanied by a beep and the software automatically fixes it in the evaluation form if you pressed on
To go to the next measurement you need to place the measuring electrode on the examinee’s skin in accordance with the next point’s illustration.
If you press the push-button before the expiration of 3 seconds from the moment of the first pressing, or there was a mistake you can press the button (back) and shift to a necessary point.
To repeat this or any other point, press the button (forward) - to pass to the next point
or the button – to shift to a previous point.
Shifting back and forth you can begin testing with a point which is displayed in the program.
This function helps you in erroneous data input.
The button (clear) – deletes all measured data during examination.
The software automatically saves 24 values and creates a test summary in the shape of different charts in the clients file.
If the active/passive probe does not start reading right away make sure that the cotton swab is moistened enough and the probe is connected to the device.
A stable running red line shows that the measurement is accurate but there can be a break in the graphic line, continue to measure a point and wait for a while, it is probably passing. A break indicates that the signal has broken. It often happens if the skin conductance is not sufficient and your energy level is low, then the measurement won’t be accurate as it won’t show the information about your health.
This section contains the graphs with examination summary for 5 elements or 6 Ki, enter your notes, complaints of a patient and post test results on a forum to get an advise from other forum members if you experience any difficulties in results interpreting.
Once you have completed points’ examination press the button to view the graphical summaries.
The user can view which meridians are out of their physiologic balance (if the meridians are deficient or excessive) in the five different style graphs.
You can view and save unlimited quantity of tests for each client in the software’s database.
Results may be viewed in five different graph styles:
The first graph shows both sides of the same meridian.
The second graph style displays twelve meridians’ energy of the left side and their pairs on the right.
Third graph allows a user to choose several previous exam results from client’s history to compare with each other over a span of time to monitor the dynamics (or the current test with previous examinations).
To do so choose the corresponding graph style (third style) , click on the dates you want to view and the program will display each test in its own color.
The advantage of this function allows you to compare the last examination with the previous ones and view any changed physiological responses.
With the help of the fourth graph you are enabled to see if there a split energy between the left and right sides of the same meridian and the level of such imbalance between the 2 sides of the same meridian.
To do so, choose the fourth graph, then click on the date you want to view (if it is not a current test).
The tab Complaints
The section enables a user to enter any notes about patient, complaints, symptoms, previous treatment, etc.
It can be done in a following manner:
Press any color you want on the color panel
with the left click of the mouse and click it on the area you want to mark.
Enter the name of the complaint you are going to describe, choose the date
and make your comments in the field below
Once you have completed describing the complaints, press the button
If you want to delete a particular complaint, choose it from the list and press the button
If you want to delete the all complaints from the list, press the button
In a need to delete a particular date with the complaints, press the buttonwith a drop down list with dates and choose necessary one and press
Automatically directs you to the forum submitting form.
This section helps a user to post a thread on our practitioner's forum directly from the software. If you have any questions concerning the examination results, how to interpret them or you want to enhance your treatment with other ideas, so you can submit a message on the forum with the test results and description of the health-issue. We respect your patient's privacy, so personal details are not disclosed, like name, last name. If you wish you can mention about gender, age and some other things which you think relevant.
If you want to post somebody's measurement results, you need first to choose a client from the client list, his personal details are not shared when posting to the forum, but his name and last name appears in the list for your reference.
Name a message your thread will be posted, choose the method and language section you want your readers to find your message (for example, if you choose English language, your message will be posted on English section of the forum).
The field with problem description will be taken from Complaints tab, but it is still possible to add some more information, as well pressing the Receipt tab enter probable description of your treatment formula you think to adhere in the field below.
Having completed entering the description of prospective acupuncture formula and questions remained with it, press the button
Since the device supplied with the software Su Jok 6 Ki has an inbuilt light in its detecting probe for colour therapy, a user is enabled to set the colours in the software, their time of generating and choose a frequency from 0 to 50 Hz the colour will be generated.
If you use the software and colour therapy function for the first time, you should create a template first and mark the points on corresponding zones of palms that you are going to influence with the colour.
Press the button
Once the software opened the colour therapy template, start marking the points on the corresponding zones of hands that you are going to treat with colour.
- a new line (task) for color correction
Preset the time of correction, colour and generating frequency (it can be chosen from 0 Hz to 50 Hz)
- adding a new line (task) for frequency correction.
With the left click of the mouse press on a correspondence zone to mark a point.
On the left window you will see the colour schedule where you should set the colour, time and frequency of generating.
You are able to create any colour from the colour palette. To do so, click on the colour line with the left mouse click and then right click of the mouse will help you to choose necessary colour from the palette.
The left part of the colour scheduling field supposed for choosing the colour, a frequency with which colour will be oscillated and time of generation.
The buttons below help a user to manage colour therapy process, starting, pausing and stopping it, as well as showing the time left for generating.
The right part of the colour scheduling field displays an order of the points (for your convenience you can mark them with different colour by the right click of the mouse).
These buttons will be also helpful in managing the points, deleting them.
NOTES for COLOR THERAPY
Color therapy can be applied in several ways described below:
The first way: we advise to choose a color the most favorite for a patient, set the time of generating and play it through the device's detecting probe (the light is inbuilt there).
The second way: select the worst 1 or 2 meridians based on test results provided by the software. Start influencing biologically active points on chosen meridians or reflexology zones (the points or zones are selected by a practitioner individually starting from the most distal points or in correspondence with a patient's health condition: disease, symptoms).
The time of color generating on one point is from 1 to 3-4 minutes depending on each patient condition. It is advised that a practitioner starts from minimal time, 1 minute, gradually making the time of generation 3-4 minutes; as well as the time of generating should be less for kids at the initial stage of therapy. One session color therapy should involve 4-6 acupuncture points. The color therapy consists of 5-10 sessions depending on treatment dynamics (you can check and monitor it with the help of Ryodoraku testing).
The third way:
If there is
no positive feedback from treatment after 1-3 sessions, try using
complementary colors in relation to the original one, based on pairs of
complementary colors: red, green, orange, blue, yellow-purple
Set the color in accordance with individual preferences of a patient or conduct color therapy session like in the second way.
The fourth way:
If there is no effect from using complementary colors shift to organ related colors:
red - Heart (Small intestine)
orange - Bladder (Kidneys),
yellow – Stomach (Spleen),
green - Gall bladder (Liver),
blue – Lungs (Large intestine)
violet – Pericardium (Triple heater).
Set the color in accordance with individual preferences of a patient
And repeat steps from the second way.
Note: in a rare case if all these steps do not show positive feedback we would like to advise to shift to another treatment modality.
Since the software provides 5 elements and 6 Ki testing, you can print measurement results for 24 points or 144 points correspondingly.
You can adjust the print settings to be printed out or choose a template from the template’s list. If you wish to print the template, it should be created first and can be used subsequently.
Setting up the Print Options.
If there were any templates created and saved previously in the software, choose any you want to print or use default template checking boxes with necessary print features.
In order to open an empty template choose Default from the template’s list or choose an existing template from the drop down list.
Check the boxes on the right with information you want to print.
For client details:
Name, last name – Name and last name of the client will be printed out.
Date of birth – if you check this box, client’s birth date will be printed out.
Registration date – the date when you registered a client’s card will be printed
Date of exam – date and time of testing.
Choosing this setting you open the list with all tests for the given client saved in a software’s database.
Graphs: choose the graph type to print.
The next print setting is supposed for whether you want to print Recommendations section or not. If you want to have in your print symptoms, notes on treatment done and your recommendations' suggestions, check the corresponding or all boxes.
Treatment - prints out the treatment recommendations you saved in the software before
Recommendations – shows treatment recommendations. Checking this setting, you open the list with the all recommendations you have entered for a certain exam results and patient.
You are able to choose previously saved recommendations or create new ones. To create a new recommendation, use the tab located in the upper right window corner.
Press the button . Enter the recommendation name in a field supposed for that.
Once you complete entering recommendation name, press the button
Write your recommendations in the field below and press the button
If you want to delete the recommendation you have saved, press button, but choose necessary one from the list prior to that.
Let us come back to the print settings which are located on the left side of the print window.
Having created all your recommendations, check necessary ones on the panel located from the left in case if you want any recommendations to be printed out in your testing report.
The software allows to create print out template with the information you want to appear on the paper.
Header and Footer for the template, use the tab located in the upper part of the software’s window.
You can also make your own order of information print out in the report. Use the panel which is in the lower part of the software’s window.
Mark one of the lines or move it with the help of UP or DOWN arrows.
Template list which is at the bottom of the right side displays the all templates saved in the software and you can easily manage them by creating a new one, saving in the software and deleting it.
Having set the all print out parameters, press the button to preview it and make changes if necessary.
To use previously created template for print out, choose a necessary one from the templates’ list and all printout parameters will be displayed automatically in the window. It is possible to create your own template for the print out. Press the button . Enter the name for a new template in the line from the right side and set all necessary print out characteristics. Press the button located in the right lower corner of the window. The new template will be displayed in the list.
Other buttons of the window Print:
- saving of the all changes in a
- printing out without saving changes in a template.
- removal of a chosen template.