The device is made of hard plastic with an inbuilt metal nozzle for testing, color generation and influencing on the point with a frequency.
There is a removable lid with a li-ion battery under it. Back cover with 2 jacks for connecting wires and a small hole for a LED that indicates the status of the device (turned off, working or pending connection)
The basic version of the device includes 2 tips (one for color generating, 1-2 mm diameter, usually working on the auricular or byol meridians of su jok.
The second tip, a spring-like with a limiter of the pressure to observe the same loading on the points, usually applied in the EAV method but can be used for Su Jok or auricular points testing), passive electrode and USB cable for connecting to the device.
Basic configuration: 2 tips, 1 electrode and USB cable, 2 electrodes and doubled wire (used for frequency therapy)
For EAV: Metal honey comb for placing vials, remedies, etc. and a wire for connection to the passive electrode
The device comes with one accumulator by default, which allows you to work more than 3 hours in continuous mode.
Turn on the device by pressing the button on the device itself, after connecting it, the device will generate a sound signal. If there is no sound when you pressed the button, then the battery is dead, and you need to charge it.
The device turns off automatically if it is in an idle state more than 3 minutes. If you want to tur off or restart the device, it is necessary to press the button and hold for 5 seconds. After this time, the device turns off.
To recharge the battery, you need to connect the wire to the device, in its dedicated connector and to the USB connector of the computer or to the charger for smartphone with 5V output. Battery charging time takes about 1 hour.
Connecting the device
The device has 2 connection variants. The first variant is through USB cable and the second variant is wireless through Bluetooth.
USB connection should be used if the program works only via USB cable or to charge the battery. It is also used to upgrade the software (if you bought one software and then after some time, you decided to buy another software, and you will need to get an access to another purchased software without sending the device back to us)
1. Port for USB connect
2. Port for passive/ frequency therapy electrode
Insert one end of the cable into the device and the other into the USB connector of the computer.
Turn on the device (press the button), run the driver (supplied with programs) to install the device on the PC.
After successful installation, run the program you are going to work with, and after a successful connection, you will hear
sound signal of the device.
Bluetooth can be used if the program works through this variant of the connection, usually it is used to work with phones, tablets as well as PC, if
the software supports this kind of connection to the device (these software are: EAV v.2.1, TCM pro v.1.0 for example).
To connect Bluetooth connection, you need to press the button on the device (the device beeps), turn on Bluetooth on the PC or phone, and press "Device search". After a while you will see a new device detected, it should be "altmed" and the device number next to it.
Make pairing of the device to Bluetooth, it may ask for a connection code (use 0000 or 1234). After some time, you will see that the device is found and connected. Further, run the mobile application on your phone/tablet or software on a PC,
the device will connect and notify you with the sound signal.
Download the program APP ( Android )for BioPulse PRO and install it. (Click Ryodoraku BioPulse)
Or open Google market on your smartphone or tablet, enter the name biopulse, find the program and install it.
If you could not connect to the device, please contact our technical support.
+34 608 920 820
Automated testing and treatment mode
Working with testing modes for acupuncture point measurements.
The automatic testing and treatment allows a practitioner to work on reflexological zones on hands and feet, where the device automatically measures the values,
analyzes measurement results and then generates colour to correct imbalanced zones.
For example, if the measurement value taken from the zone is imbalanced, then a
healing colour is generated. If the value is within the normal range, the device does not generate any colour.
To determine the measurement value from the selected zone correctly, it is necessary to hold the passive electrode
in one hand (for self testing or, if you test a client, then the passive electrode is held in his hand); with the
other hand holding Biopulse Pro, you search the zones for imbalanced electro-conductivity. The zone you are searching
with Biopulse Pro should be moistened with water or saline solution.
Color therapy mode
If you want to generate different colours (8 colours total) with the help of a device like a colour pen, then press the middle button 2 times, and you will jump to colour pen mode.
After pressing the button, select the necessary colour and work with your desired colour over the zones for 1 minute. If you want to generate the same colour again, you should press the button again after the previous time was completed.
If you press the button while certain colours are being generated, the device will start to generate the next colour.
This way you can select between the desired colours to be generated.